Series 5000 - Near Infrared Photometers
Where can I find the 4-20 mA output modules on the computer motherboard?
The 4-20ma modules are supplied on of two ways:
as a component from Analog Device, Part Number 1b-22 AN
(device with the white label) or
as a separate circuit board mounted on the top.
Top of page
When I try to calibrate analyzers by pressing Zero and span it is showing "This function is password protected". Please advise how to remove it.
I suspect someone pressed the logout function. First they should enter the password function and see if the default password is there and press enter to log back in. My hope is that nobody changed the password and left the default password. If this does not work, then yes, the only way to go is cold start. Press the system button and power up to go back to factory default. But before that copy down the pressure compensation factor. Then after cold start, re-enter the pressure compensation factor and re-do zero and span.
Top of page
Document Control & Training (password)
e-mail us: email@example.com
Copyright © 2010 Teledyne Technologies Incorporated. All rights reserved.
Teledyne Analytical Instruments 16830 Chestnut St City of Industry, CA 91748 USA